Question about Measuring Tools & Sensors
If you select current measuring (A or mA) and have the test lead connected to V then you get this alarm. Same thing if you select Voltage measuring and the lead is connected to a current terminal.
Check the connections and the selector setting.
Posted on Feb 10, 2019
Fluke 88 V displays "LEADS" warning if test leads are in A or mA/uA terminals but switch position does not corespond with those terminals.
Remove all leads and if you still get "LEADS" warning try cleaning all terminals with air. Sometimes dirt gets in A terminals and unit thinks that leads are connected.
Use compressed air or airduster to blow dust from all terminals and hopefully that will fix the problem.
Posted on Jan 08, 2012
Tips for a great answer:
Jul 19, 2013 | Measuring Tools & Sensors
Jun 19, 2013 | Fisher and Paykel IWL12 Top Load Washer
Jan 24, 2013 | Fluke Networks Fluke 87-5 Digital...
12 V lead-acid rechargeable battery.
Dec 26, 2012 | Fluke 1550C Insulation Tester,battery,50...
Mar 18, 2012 | Fluke Networks Fluke 179 Electricians...
1,030 people viewed this question
Usually answered in minutes!