20 Most Recent
Mitutoyo 500-672 0-6" ABSOLUTE Digimatic Coolant Proof Inch/Metric Caliper Questions & Answers
Mitutoyo error code E552?
Error code E552 is usually caused by a communication issue between the CMM and the computer. You can try restarting the CMM and the computer, and ensuring that all cables are properly connected. If the issue persists, you may need to take your CMM to a certified Mitutoyo service center for further diagnosis and repair.
2/15/2023 6:41:53 AM •
on Feb 15, 2023
Mitutoyo error code E940?
Mitutoyo is a brand that produces precision measuring instruments such as micrometers, calipers, and height gauges. An error code E940 on a Mitutoyo instrument typically indicates a problem with the instrument's communication interface. This can be caused by a variety of issues, such as a problem with the instrument's cable or software, or a problem with the computer or device to which the instrument is connected.
To troubleshoot the issue, you should first check the connections between the instrument and the computer or device to ensure that they are properly connected and secure. If the connections appear to be fine, you may need to update the software on the instrument and computer or device to resolve the issue.
1/12/2023 8:21:54 PM •
on Jan 12, 2023
QM-Data 200 Mitutoyo with Optoeye-200. How to change or replace battery please?
Apparently the battery is Circuit Board mounted and is a backup memory battery.
Li (lithium) battery • The Li battery is mounted on the printed circuit board
in the data processing unit "QM-Data". Do not dispose of the data processing unit by burning it. If the data processing unit is thrown into fire, the battery may leak chemicals, burst, or burn violently. Comply with government regulations regarding the disposal method.
• This error message indicates that the memory backup battery is low, and that if you continue to use the data processing unit "QM-Data", backup will not be performed. Save the part programs and measured data stored in the internal memory of the data processing unit to the floppy disk or to a personal computer. • Contact your supplier to replace the battery.
It also says changing the battery removes any saved data.
So if you are looking to do it yourself, look for screws to get inside.
1/29/2022 11:25:16 PM •
on Jan 29, 2022
How to correct an e671 error message
I realise this question is old, but you never know, might help someone in the future. This particular error is that the probe is not correctly seated, usually the magnets have become slightly loose. The solution is to take the probe head off the cmm, and push the magnets against your surface table or suitable flat surface, this should push the magnets into their correct position. If this doesn't help, it may be that you've managed to break that particular part, if you have a probe rack, try a different probe. Lastly, if you crashed the CMM by hitting something, ensure you didn't bend your probe in the process (and the ruby is still there!), this doesn't help the error code fix, but worth checking as the cmm crashing into something at speed is the most likely cause of this E671 error.
11/24/2021 3:25:06 PM •
on Nov 24, 2021
I have a mitutoyo 500-422 and how do I replace the battery
I also have a Mitutoyo 500-422 Caliper. But mine is strictly solar powered. There is no battery for it. So long as you have 60 lux of illumination on it, it will work fine. Its not suited for darker environments, as it will lose its measurement when the light falls below 60 lux.
10/16/2019 5:07:49 PM •
on Oct 16, 2019
What does Mitutoyo error code E380 mean?
These 2 signals are drive prohibition (overtravel) inputs in forward and reverse directions. If Drive Prohibition Input Setting (Pn504) is 1, you can use the setting of Stop Selection (Pn505) to select the operation to be taken upon input of each prohibit signal. If Drive Prohibition Input Setting (Pn504) is 2, drive prohibition input protection (E380) actuates upon input of a drive prohibition.
3/2/2019 9:40:00 AM •
on Mar 02, 2019
Not finding what you are looking for?