20 Most Recent
Mettler Toledo - RM40 - LiquiPhysics Excellence Benchtop Questions & Answers
MANUAL OF RM40 REFRACTOMETER
Hello, Need Service Manual Refractometer Mettler Toledo RM40, thats problem tsurus.exe while start up Problem,
Please can I have assistance on the recalibration of my super pro miosture meter. The machine is measuring 2 % higher then it should on maize
Sure, I'd be happy to help you recalibrate your Super Pro moisture meter. Here are the steps you can follow:
- Find a reference sample: You will need a sample of maize that has a known moisture content that is different from the current reading on your moisture meter. You can either purchase a certified reference sample or make your own by drying a sample of maize in an oven or using a moisture analyzer.
- Adjust the calibration: Turn on your Super Pro moisture meter and press the "calibrate" button. Using the reference sample, adjust the calibration of the moisture meter by adding or subtracting the required percentage points to match the known moisture content of the reference sample. For example, if the reference sample has a moisture content of 14% and your moisture meter is reading 16%, you will need to subtract 2% from the calibration.
- Verify the calibration: After adjusting the calibration, test the moisture meter again using the reference sample to verify that the readings match the known moisture content. If the readings do not match, adjust the calibration again until you get an accurate reading.
- Document the calibration: Once you have verified the calibration, document the adjustments you made and the reference sample used for future reference.
It's important to note that different types of maize can have different calibration requirements, so it's a good idea to recalibrate your moisture meter regularly and for different types of maize samples. You can also refer to the manufacturer's instructions for specific calibration procedures for your Super Pro moisture meter.
My fluke 21 III doesn't read a/c or d/c voltage properly
If your Fluke 21 III multimeter is not reading AC or DC voltage properly and the ohms sign comes on in voltage mode, it may indicate a problem with the internal circuits of the multimeter or the leads. Here are some steps you can take to troubleshoot the issue:
- Check the leads: Ensure that the leads are properly connected to the multimeter and the test points. Make sure there are no breaks or cuts in the leads. Try using a different set of leads to see if the issue persists.
- Check the batteries: A weak or dead battery can affect the accuracy of your multimeter readings. Check the batteries to ensure they are properly installed and have sufficient charge.
- Check the settings: Make sure that you have selected the appropriate range and function on the multimeter for the measurement you are trying to take.
- Reset the multimeter: Reset the multimeter to its default settings to see if this resolves the issue.
If none of the above steps resolve the issue, it may indicate a problem with the internal circuits of the multimeter, and it may need to be repaired or replaced.
Wiring diagram for 96 peterbilt 379 volt meter
Here is a possible wiring diagram for the voltmeter on a 1996 Peterbilt 379 with a C15 engine:
- The voltmeter gets power from the battery through a 5A fuse (F2) and a 10A circuit breaker (CB7).
- The voltmeter is grounded through a ground wire (GND) connected to the chassis.
- The voltmeter receives the signal from the alternator through a wire (ALT) connected to the alternator "sense" terminal.
- The voltmeter also receives a signal from the ignition switch through a wire (IGN) connected to the ignition switch "ACC" terminal.
Here is a possible wiring diagram based on this description:
Battery positive (+) ---[F2]---[CB7]--- Voltmeter (+) ' ' Battery ground (-) -------------[GND]--- Voltmeter (-) ' ' Alternator "sense" terminal ---[ALT]--- Voltmeter signal input ' ' Ignition switch "ACC" terminal ---[IGN]--- Voltmeter signal input
Note: This is just a possible wiring diagram and may not reflect the exact wiring configuration in your specific vehicle. It is always recommended to consult the wiring diagram provided in the vehicle's service manual or consult a qualified mechanic or electrician for assistance.
Snap-on Eco plus shows for example that 3.0 lbs transferred during ISV tank fill then immediately shows the tank having 1 lb 9oz what is wrong?
Based on the information provided, it's difficult to determine exactly what is causing the discrepancy between the amount of refrigerant transferred and the amount of refrigerant shown in the tank. However, here are a few potential explanations:
- Measurement error: It's possible that there was an error in measuring the amount of refrigerant transferred or the amount of refrigerant in the tank. For example, if the scale used to measure the refrigerant is not calibrated correctly, it could be providing inaccurate readings.
- Leaks: If there is a leak in the system, refrigerant could be escaping and not making it into the tank. This could result in a lower amount of refrigerant in the tank than expected.
- Temperature changes: The amount of refrigerant in a tank can be affected by temperature changes. If the refrigerant was transferred at a different temperature than the tank, it could result in a discrepancy in the amount of refrigerant shown in the tank.
- Software error: It's possible that there is an error in the software used to display the amount of refrigerant in the tank. This could be causing the incorrect reading.
It's important to troubleshoot the system to determine the root cause of the discrepancy and take appropriate action to address the issue.
I have a Digitech XC0436 Weather Station.
Remove the batteries from both units for a couple of hours. Replace the indoor unit battery and then the outdoor battery. This will effect a reset of sorts. Don't forget your rainfall can be different to someone 600 meters away. Having said that 100mm / 4 3/4" difference is a lot.
34788 Recover does not work
"Recover" is a term that can be used in various contexts, can you please provide more information on what you are trying to recover and what is the specific issue you are facing? If it is a software issue, please specify the software and the operating system you are using.
Also, please provide more information on what you mean by "does not work", is there an error message or any other symptom that you are experiencing?
This information will help me understand the issue better and provide a more accurate solution.
How to set electrode on pilot for empire rh50b
- Turn off the gas supply to the heater.
- Locate the pilot assembly on the heater. It is usually located near the bottom of the unit.
- Remove the pilot shield, which is typically a metal or plastic cover that protects the pilot assembly.
- Locate the electrode, which is a small metal rod that is located near the pilot burner.
- Adjust the position of the electrode so that it is directly in front of the pilot burner, with a distance of 1/8 inch to 3/16 inch between the electrode tip and the pilot flame.
- Replace the pilot shield and turn the gas supply back on.
- Test the pilot flame to ensure that it is stable and blue in color.
My figure has DigiTech XC-0366 wireless weather station is not collecting the outdoor temperature
Normally it has to be done in order. Remove both lots of batteries. Put the base unit batteries in first and then put in the outside unit batteries. they should pair. In fact the manual states that.
- Open the battery compartment on the weather station and insert 2 x AA
batteries, taking care to put them in the correct polarity. Close the battery
- Open the battery compartment on the transmitter and insert 2 x AA batteries,
taking care to put them in the correct polarity. Close the battery
Note: Ensure to install the batteries into the weather station and transmitter
within the same hour to ensure that they synchronize properly.
DigiTech XC-0366 User Manual - ManualsLib
https://www.manualslib.com > ... > Weather Station
View and Download DigiTech XC-0366 user manual online. Wireless Weather Station With Outdoor Sensor. XC-0366 weather station pdf manual download.
Sokkia Total sation getting the error message E123 & E124. What is the problem.
The error messages E123 and E124 on a Sokkia total station typically indicate a problem with the instrument's distance measuring system. This could be caused by a number of issues, including problems with the instrument's internal hardware, issues with the reflector being used, or problems with the environment in which the instrument is being used.
To troubleshoot this issue, you could try the following steps:
- Check the instrument's settings and make sure that it is configured correctly.
- Check the reflector being used and make sure that it is clean and properly aligned.
- Make sure that the instrument is being used in an appropriate environment. For example, it should not be used in extreme temperatures, high humidity, or in the presence of strong electromagnetic interference.
Not finding what you are looking for?