Question about Pride New Jazzy 600 Power Wheelchair by Mobility
When charging battery, no charging indicator comes on, and still 3 red power LEDs flashing rapidly continously.
The control systems on power mobility equipment have a "prevent" circuit that immobilizes the equipment while it is plugged into a charger, so folks won't take off with it plugged into the wall outlet. When you turn it on when plugged into a charger, the display will flash constantly, or sweep constantly, depending on the manufacturer. This is normal. Your off-board charger should have a red light to indicate power to the charger, and another light for the charge status. Yellowish means it is charging, when it turns green, you are fully charged. If the status light is green all the time, that is an indication that the charger is not functioning properly, or that the batteries are below a chargeable state. Check charger output, which should be 26-28 volts dc, and each battery should show at least a 12 volt reading, but around 13.5 volts with a full charge. If one or both of the batteries drops below 12 volts after a short period of time, it's time for a fresh pair. Fully charge new ones before using the chair.
Posted on Jul 12, 2009
Tips for a great answer:
Oct 05, 2017 | Pride Health & Beauty
Sep 25, 2013 | Computers & Internet
Sep 25, 2013 | Gateway Computers & Internet
Mar 04, 2013 | Canon VIXIA HF M41 Camcorder
Sep 21, 2011 | Plantronics Explorer 220 Bluetooth Headset...
Jun 30, 2011 | Compaq Presario Notebook
Aug 12, 2009 | Intec PlayStation 3 Turbo Shock III...
Jul 12, 2009 | Intec PlayStation 3 Turbo Shock III...
Jan 27, 2009 | Dell Latitude D600 Notebook
Sep 23, 2008 | Computers & Internet
581 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: