Question about Science & Laboratory
The fault table is on page 15 of the following document.
Further support may be channeled through the distributor. As with many devices, it is suggested to have mainteance and repair done by personnel trained specifically in such for the devices. There is quite a bit of information in this manual one might wade through Page 28 lists error codes and error 2 listed refers to door open during cycle.
UK Customers Eschmann Equipment, Peter Road, Lancing, West Sussex BN15 8TJ, England. Tel: +44 (0) 1903 765040. Fax: +44 (0) 1903 875711. Overseas Customers Contact your local distributor. In case of doubt contact Eschmann Equipment.
Posted on Jun 19, 2019
SOURCE: My autoclave ses 2000 turns
I would say that there is a problem with the thermostat. There is a small relay that pulls in to heat. And the relay opens when the thermostat reaches desired temp. I say it is the thermostat because it is pulling in the relay and heating. And if it is going to 107 degrees every time then the thermostat is pulling in the relay. So that tells you that the relay is working properly. So it would have to be the thermostat.
Posted on Sep 14, 2010
Tips for a great answer:
Mar 06, 2018 | Science & Laboratory
Aug 31, 2017 | Science & Laboratory
Nov 29, 2014 | Cars & Trucks
May 06, 2014 | Tuttnauer EZ10 Automatic...
Apr 16, 2014 | ESCHMANN Little Sister Autoclave Tabletop
Jan 29, 2014 | ESCHMANN Little Sister Autoclave Tabletop
Feb 23, 2011 | Exercise & Fitness
Jan 14, 2010 | Medical
Sep 29, 2017 | Science & Laboratory
198 people viewed this question
Usually answered in minutes!