Question about In Office Equipment & Supplies
Transformer lead broken / not connected Connect lead or replace transformer You will need to return the MIT520/2 to Megger for Repair.
Posted on Apr 03, 2018
Tips for a great answer:
Jan 28, 2016 | Megger MIT510/2 5 kV Insulation Resistance...
Mar 09, 2015 | Megger MiT515 5kV Insulation Tester...
Jan 20, 2015 | Measuring Tools & Sensors
Unit produced E78 on start-up. Transformer lead broken / not connected Connect lead or replace transformer
Incompatible measurement board software Change measurement board
Unit produced E78 at the end of a test IrDA screen missing from one board Change board with missing screen.
May 25, 2014 | Megger MIT1020/2 Megohmmeter, Battery
Jul 28, 2011 | RTO Megger 10 Kv Insulation Resistance...
1,214 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: