Question about Select Jazzy Used Electric Wheelchair
It's likely after sitting for so long that the batteries are shot. Sealed-gel-batteries should be fully charged, then totally disconnected before storage, and then put back into service within a year, or placed back into the chair and recharged again, and disconnected again.
Test the batteries using a voltmeter set to DCV, in the 20 volt or more range. Put the probes across the terminals. A good 12 volt battery should read in the 12.5-13.5 voltage area. If it isn't hitting those numbers after a charge, it is no good; time to recycle. Your chair takes two batteries, which are gel, Group 22 or Group 24 type and will cost you about $350-$450 for a set. Group 24 will give you the maximum range for this chair.
Posted on Jul 07, 2009
Tips for a great answer:
Jul 15, 2014 | Electric Mobility Invacare Pronto M41...
Jun 13, 2014 | Invacare Pronto M51 Power Wheelchair
Mar 21, 2014 | Invacare Pronto M51 Power Wheelchair
Jun 19, 2013 | Pride Jazzy Select 6 Indoor Power...
Jun 20, 2012 | Pride Jazzy 1103 Ultra Standard Power...
Apr 04, 2011 | Pride Mobility Products Health & Beauty
May 15, 2010 | Invacare Supply Group Invacare Pronto M91...
Aug 14, 2009 | Invacare Pronto M51 Power Wheelchair
Aug 08, 2009 | Invacare Supply Group Invacare Pronto M51...
435 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: