Question about Invacare Pronto M51 Power Wheelchair
Where can I find a invacare m51 electronics manual
I suggest you visit http://www.invacare.com/doc_files/1125085.pdf for further information.
Best of luck! Handicap Wheelchair Vans Hand Control For Cars Mobility Scooter Better...
Posted on Aug 27, 2014
SOURCE: oil / gel leak
If it is coming from your motors, it could be very bad. i.e. motor replacement. Sometimes on the side of those motors there is a black plastic cap about the size of a quarter or half dollar, those fall out sometimes and allow gearbox grease to come out. This should`nt be confused with your motor brush caps which are about the size of a pea and have a small **** in them for a screwdriver.DO NOT TAKE THESE OUT. If the large cap has come out just stick it back in. If you can`t find it you`ll ned to call a dealer to get the part replaced. If you can still find no prob,your probably going to have to go to a dealer.
Posted on Mar 26, 2009
SOURCE: speed control knob has come off
you need Invacare part # 1040225, cost about $6.00. Call these people 866-722-4581
Posted on Sep 22, 2009
Testimonial: "I looked on the internet for this part for HOURS without success. The "FixYa" expert found the model number needed and the phone # to order. Thanks!! "
The 9 flash error code indicates a communications fault. The communication fault relates to the communication between the joystick and the control module. The joystick on the arm mount is connected to the control module in the base. When there is a break in the wiring of this wiring harness, also called a bus cable, a fault occurs. This trips the computer and the computer sends out a flash code to the joystick. Simply stated there is a broken or pinched wire between the joystick and controller. Look physically at the wire and check for cuts, breaks or pinches. The best short term repair is to repair the broken wire with solder and heat shrink or tape. You can also use **** splice connectors and tape. The best long term repair is to replace the harness with a part available from Invacare. The most definitive way to diagnose this problem is to read the fault log of the controller using a hand held programmer available to dealers and certified technicians. The number and type of fault codes logged in the controller is another indicator of which
component(s) has failed. Hope this helps.
Posted on Nov 20, 2010
If you have had swing away footrests on this chair you can replace them with model PHW93 and must ask to get them in red when you order them from a wheelchair store. If you dont know where a store is that sells Invacare products, call Invacare customer service at 800-333-6900.
Posted on Apr 07, 2011
Have you had the batteries load tested? How do you know the batteries are good?
Posted on Apr 21, 2013
Testimonial: "Took the batteries back to Batteries Plus and had them tests. Load test was good but the capacity test showed batteries should be replaced. Replaced batteries but chair drives slower then it did with the bad batteries....barely moves when on low setting. "
Tips for a great answer:
Jan 12, 2018 | Invacare Health & Beauty
Dec 25, 2015 | Health & Beauty
Aug 24, 2014 | Invacare Pronto M51 Power Wheelchair
May 06, 2014 | Jet 3 Ultra Power Chair
Jan 29, 2011 | Invacare Pronto M51 Power Wheelchair
Oct 11, 2010 | Invacare Pronto M51 Power Wheelchair
Oct 04, 2009 | Invacare Pronto M51 Power Wheelchair
787 people viewed this question
Usually answered in minutes!