Question about Zeiss Humphrey 740 Visual Field Analyzer
Aperture may be misaligned
I believe "state 19" is a weak main illumination bulb. Not sure if it's the same on your 740, but that's what the problem was with our 750i. Even though the bulb appears to be working, they dim over time and won't pass the start up calibration checks. On ours, there is a round access panel on top of the unit to service this bulb. Just be careful removing the bulb and don't bump any of the clear optical wheels. Part # 2660021106082 (for 750i. not sure about 740) Call Zeiss to confirm. Realize this is an old post so info is for future readers.
Posted on Jul 27, 2015
Tips for a great answer:
Jun 07, 2018 | Zeiss Science & Laboratory
Nov 04, 2014 | Zeiss Humphrey Acuitus 5015 Autorefractor...
Aug 04, 2014 | Zeiss Humphrey 740 Visual Field Analyzer
May 24, 2014 | Zeiss Humphrey 740 Visual Field Analyzer
Mar 21, 2014 | Zeiss Humphrey Acuitus 5015 Autorefractor...
Aug 26, 2013 | Zeiss Humphrey Acuitus 5015 Autorefractor...
Mar 29, 2013 | Zeiss Humphrey Atlas Eclipse 992 Corneal...
635 people viewed this question
Usually answered in minutes!