- If you need clarification, ask it in the comment box above.
- Better answers use proper spelling and grammar.
- Provide details, support with references or personal experience.
Tell us some more! Your answer needs to include more details to help people.You can't post answers that contain an email address.Please enter a valid email address.The email address entered is already associated to an account.Login to postPlease use English characters only.
Tip: The max point reward for answering a question is 15.
When the at2 is in auto mode, first of all it check the leads are properly connected or not.Then it will acquire the ECG and store the same in its memory then only it starts printing. that's why it takes that much time to give a printout. When its starts printing you can remove the connections from the patient because it have alreadymemorized the ECG.
If interpretation is in on position it will compare the acquired ecg with the standards in its memory (allready programed by the manufacturer). and will give a detailed report with ecg.
The heat on the print head can fade over time giving you a poor print. also the feed roller can wear and give poor pressure to the print head. thw way to diagnose it is to put a peice of electrical tape on the print roller and print an ekg. if it prints well in the area where the tape was, it is the print roller. if there is no change it is the print head.
Does it work in auto mode ? are you putting paper correctly means , there is a black mark on paper which is sensed by paper sensor .If you put paper up side down then paper sensor won't be able to sense any paper & gives you error for rifilling paper .If this doesn't help you then You need to replace battrey . Thanks
If it stops printing completely in the middle of doing the print, it is because the battery is bad and needs to be replaced. If it only prints some of the leads but it does print the full report it is because you have the leads mixed up (Wrong spot on the patient) or the leads are mislabeled or some of the clips or electrodes are bad. I can fix these. Let me know if you need help.