Question about Oregon Scientific Measuring Tools & Sensors
Remote Temp/RH sensor THGR928N works fine for 2 days after which no signal received on weather station WMR928N. Batteries changed twice. Did long reset of both sender and receiver twice. Problem remains. Inside is clean with no corrosion. Is there an alternative compatible unit? Regards, Chris (UK) 24 Mar 13
I found this in the manual, simply pressing the RESET button won't start communication again. Try this:
Your main unit displays current, minimum, and
maximum temperature and humidity readings
collected by the remote sensor and main unit.
Outdoor data is collected and displayed every
40 seconds and indoor data is collected and
displayed every 10 seconds.
The red LED flashes whenever the sensor
transmits an RF signal to the receiver.
select sensor or use auto scan
• Press CHANNEL to switch between
• To auto-scan between sensors, press and hold
CHANNEL for 2 seconds. Each sensor's data
will be displayed for 3 seconds. To end autoscan,
press CHANNEL or MEMORY.
• Press and hold CHANNEL and MEMORY
simultaneously to search for remote sensors.
resetting the system
The RESET buttons are located on the back of the
main unit and sensor. Use a pointed subject to
press RESET when you change the batteries and
whenever the system malfunctions.
Posted on Jul 13, 2011
You must move the remote sensor close to the base unit to reset it and it will work fine. After it shows the temp then you can move it back to its location. thanks for asking at Fixya.com.
Posted on Dec 02, 2010
Tips for a great answer:
Jan 07, 2016 | Measuring Tools & Sensors
Jan 05, 2016 | Taylor Measuring Tools & Sensors
Dec 28, 2015 | Measuring Tools & Sensors
Dec 16, 2015 | Audio Players & Recorders
Mar 08, 2015 | Measuring Tools & Sensors
Nov 04, 2013 | La Crosse Technology La Crosse S98642...
Feb 27, 2012 | Measuring Tools & Sensors
Sep 11, 2010 | Honeywell TE852W Weather Forecaster With...
Feb 21, 2008 | Computers & Internet
1,739 people viewed this question
Usually answered in minutes!