Question about Microtek SCANMAKER 4850 Flatbed Scanner
I am trying to clean the glass on a Microtek 4850. I need to know how to open the case. The Microtek support downloads do not contain the information for that chassis model. Note that I have opened the upper end, but cannot see how the lower end is connected. I have tried pushing the upper end off as well as gently prying the lower end. Both without success so far.
Opening a Tevion(Microtek)3200: thanks for the paperclip tip! That gets us in. Then the straightened wire coat-hanger slipped through the front corners finishes the job. The clips are on the base this time, about 2.5 inches from the corners set in about half an inch from the joint. They hook into loops on the top panel and have to be levered the opposite way to the paper-clip ones, that is, towards the joint. The top still remains attached by the ribbon cable,I undid the 4 self-tappers affixing the front switch plate,also maybe as well to lock the mechanism before you start: turn the little orange screw underneath with a 20p piece or a dime or ten centimes (I'm sorry, I don't know the sterling /yuan exchange rate) ,Happy New Year, Redpath
Posted on Jan 04, 2008
These things are made not to be opened. As they are sealed in manufacture I cannot see why you should want to clean the glass from the inside.
Posted on Jun 07, 2007
Tips for a great answer:
May 18, 2010 | Microtek SCANMAKER 4850 Flatbed Scanner
Apr 11, 2010 | Microtek SCANMAKER 4850 Flatbed Scanner
Nov 29, 2009 | Microtek SCANMAKER 4850 Flatbed Scanner
Feb 12, 2009 | Microtek ScanMaker 9800XL Flatbed Scanner
Jan 30, 2009 | Microtek SCANMAKER 4850 Flatbed Scanner
Feb 18, 2008 | Microsoft Windows XP Professional for PC
Feb 09, 2008 | Microtek ScanMaker 3600 Flatbed Scanner
Usually answered in minutes!
Step 2: Please assign your manual to a product: