Question about Health & Beauty
I just bought a used Invacare Pronto SureStep M91 with Elevating Seat that is a VA chair. The batteries were ignored when the owner passed away. They also lost the owner's manual and I can NOT find out how to remove the seat so I can access the batteries and install new ones. How do I remove the van seat from this chair?
Single post, lift handle to rotate then pull up very hard!!! four poster lock is between knees release tilt back and remove
Posted on Oct 31, 2012
The 9 flash error code indicates a communications fault. The communication fault relates to the communication between the joystick and the control module. The joystick on the arm mount is connected to the control module in the base. When there is a break in the wiring of this wiring harness, also called a bus cable, a fault occurs. This trips the computer and the computer sends out a flash code to the joystick. Simply stated there is a broken or pinched wire between the joystick and controller. Look physically at the wire and check for cuts, breaks or pinches. The best short term repair is to repair the broken wire with solder and heat shrink or tape. You can also use **** splice connectors and tape. The best long term repair is to replace the harness with a part available from Invacare. The most definitive way to diagnose this problem is to read the fault log of the controller using a hand held programmer available to dealers and certified technicians. The number and type of fault codes logged in the controller is another indicator of which
component(s) has failed. Hope this helps.
Posted on Nov 20, 2010
If you have had swing away footrests on this chair you can replace them with model PHW93 and must ask to get them in red when you order them from a wheelchair store. If you dont know where a store is that sells Invacare products, call Invacare customer service at 800-333-6900.
Posted on Apr 07, 2011
Tips for a great answer:
Sep 29, 2015 | Invacare Pronto M91 with SureStep
May 28, 2014 | Electric Mobility Invacare Pronto M41...
Mar 31, 2014 | Invacare Pronto M51 Power Wheelchair
Apr 02, 2013 | Invacare Pronto M91 with SureStep
Sep 16, 2011 | Invacare Supply Group Pronto M61 Standard...
Jul 18, 2010 | Invacare Supply Group Invacare Pronto M91...
May 15, 2010 | Invacare Supply Group Invacare Pronto M91...
Nov 05, 2009 | Invacare Pronto M51 Power Wheelchair
274 people viewed this question
Usually answered in minutes!