Question about GE Marquette Eagle 4000 Vital Signs Patient Monitor with Ecg Nibp SPO2
There will be an MOV (metal oxide varistor) After fuse. This MOV is rated 125v. When you plug the machine in to 220v supply this MOV will become shorted and its blows the fuse to prevent further dameges to the power supply. Replace the MOV and then insert the fuse. I have added the image of MOV here. It is placed parallel to the AC mains supply rail after fuse.
Posted on May 07, 2012
Tips for a great answer:
Oct 20, 2014 | Jenn-Air Refrigerators
Jan 29, 2011 | LG LS-K1830HL Heat Cool Mini Split Air...
Dec 21, 2010 | Dell Computers & Internet
Jan 25, 2010 | Akai LCT3785TA 37 in. LCD HDTV
Aug 04, 2009 | Nintendo Wii Console
Apr 25, 2009 | Jandy AQUALINK RS TELELINK KIT
Jan 23, 2009 | Vizio P50HDTV10A 50 in. Plasma HDTV
Apr 09, 2008 | Canon - SUPER G3 FAX BOARD-T1 (0450B002AA)
217 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: