Question about Optics
About a 6 year old Meade LX-200 telescope. Keeps blowing the fuse in either the DC power supply or the AC power supply when I turn it on.
One thing to check, would be the filter capacitator that is on the front circuit board. They are known to become shorted. The cap is fairly large, and may be mounted on the side of the circuit board that faces the front panel. Clip one lead, and apply power. If the fuse does not blow, then that is it. A second possible component, is the polarity protection diode, that is near the power jack, next to the fuse. If it is shorted, then the fuse will blow every time.
Posted on Aug 24, 2008
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Apr 11, 2012 | Fender Music
Oct 09, 2008 | Meade Optics
Jun 27, 2011 | Meade 114EQDH4 (36 x 114mm) Telescope
Sep 27, 2010 | Meade 8" LX90 Schmidt-Cassegrain Telescope...
May 04, 2010 | Meade Optics
Oct 27, 2009 | Meade LX90 Telescope
Oct 04, 2009 | Meade LX200GPS Telescope
Aug 09, 2009 | Meade Autostar Suite Autostar Suite...
Apr 26, 2017 | Optics
96 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: