Question about Mabis NebXP Portable Handheld Compressor
This is for the John Bunn Compact Ultrasonic Model JB0112-066. I found that the liquid inside the nebulizer ultrasonic chamber had dried up or leaked out and that was the cause of improper mist flow. You can tell if it's low on fluid by tipping the unit on its side and looking through the transparent membrane. If it's only 3/4 full or less it need fluid. If you're handy you can open up the unit by removing two screws located deep in the two holes on the bottom of the unit. Carefully pull the top off and lift out the guts. You'll see a metal plate with several little screws - undo them and lift out the rubber gasket, carefully pry out the round disc with the two wires and the rubber gasket under it. You can then use a syringe to refill the chamber with distilled water until you can see the water level about 1/4 " from the top (actually the bottom). Close everything up and test it. Hint: when putting the guts back be sure the circuit board is seated below the blue on/off button and that the two posts are coming through the circuit board. If you have any resistance when putting the top on it means the circuit board is not seated deep enough. Good luck! This will save you $75 or more for a new unit and you get lots of kudos from your spouse for being so smart !!
Posted on Jan 24, 2013
Check with medicine chamber for leakage and breakage
Posted on Jul 29, 2011
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 03, 2013 | Windchaser Outdoor Misting Fan WC161 Stand...
May 04, 2011 | Clairol Sheer Mist Aerosol Regular Hold...
Dec 25, 2010 | Fusion Beauty Glowfusion Air Glow Gun +...
Feb 09, 2010 | Vicks Humidifiers
Jan 16, 2010 | 1993 Ford Ranger SuperCab
Feb 15, 2009 | Bionaire Humidifiers
Dec 16, 2008 | Bionaire CM1 Humidifier
Dec 31, 2007 | Sunbeam Ultrasonic 81696 Humidifier
430 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: