Question about Computers & Internet
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: NCR 7875 SCANNER/SCALE
Well regarding this error sending you this display of the scanner in most cases is needed because if the scanner calibration, you have to do is:
Remove the dish from scanner (where you put the products to weigh), there you will see that has a plastic part on the right side, and take away that the bottom is a black button color you need to start to put pressure gauge.
Once you press this button, and is ready to start calibrating, then put him back to the plate and the scanner with a voice that sends you will be asking some weight, so this should have on hand your weights to calibrate the scale.
First the scanner asks 0 kg. in this step not going to put anything in weight, ahi press the button Scale Zero appearing in the panel's scanner,
After asks you to 2.5 kg, is there when you need to put the 2.5 kg asks you, when you have them placed on the plate press Scale Zero, the scanner then will ask now that you put 5kg, then you take away those who had 2.5kg and you put the 5kg, and when you have 5kg in the dish press the button Scale Zero, now the scanner will ask that you put 10k, then take away the 5kg and that he had put the 10k that asks you to press the button and Scale Zero, and finally asks you to the scanner in this 0Kg, then you take away the 10k and leave it with nothing, and press Scale Zero
If the calibration did well Scanner you send a voice message that says Calibration Complet or calbración completed, it is very important that when you calibrating the scanner is not moving or having obstacles that may make the calibration is not made well, good clean plate and that this part of the bottom, which revises the 4 Gomito that has to sustain the plate are complete, that are not broken up or are some more than others and that when you put the dish did not move. fijate also not hit the plate or the chassis of the scanner with something else, so there's no obstacles in the calibration,
If you salio calibration is not well in the display will show a 9Error, this means that there was interference during calibration, it is required to check well before the scanner or do a cleanup.
Posted on Jun 30, 2008
It's possible that the tray is pushed into max position.
Incurred due to an overload on the tray for a long time or a unprotected transport probably ...
Try turning it on, while holding the tray pulled up (eg scissors sliding under the tray to keep it elevated) and if the scale will show you "0.00", the tray is pressed...
Posted on Sep 15, 2010
Tips for a great answer:
Nov 30, 2014 | DigiWeigh 100 X 0.01 Gram Digital Jewelry...
Sep 12, 2014 | Salter 2001 electronic kitchen scale
Jan 25, 2014 | Ozeri Pro II Digital Kitchen Scale in...
Sep 05, 2012 | Royal EX330 Shipping Scale 29533D
May 28, 2011 | Kitchen Appliances - Others
Feb 16, 2011 | Avery WeighTronix Berkel WeighTronix 6710 ...
Jul 27, 2010 | Tanita 1479V - Digital Pocket Scale /...
Jun 13, 2009 | Taylor Body Fat Analyzer & Scale, Model...
Dec 18, 2008 | Taylor 7532 Ultra Thin Glass Digital Scale
147 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: