Question about Nikon Science & Laboratory
It might be that Objective lens is not seated properly or very dirty. So perhaps by rotating the objective block may help before trying to clean it. Also try to start with the lowest magnification to see that you have functional microscope before trying higher magnifications. Good luck.
Posted on Mar 21, 2011
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Jan 18, 2011 | Meade Infinity 114EQ-DH Telescope
Dec 31, 2010 | Optics
Sep 27, 2010 | C And A Scientific IExplore Scope
Sep 03, 2009 | Celestron NexStar 80GT (189 x 80mm)...
Aug 13, 2009 | Bushnell 78-9669 Telescope
Apr 18, 2009 | Celestron NexStar 130 SLT (306 x 130mm)...
Mar 25, 2009 | Vivitar (1607225) Telescope
Mar 05, 2009 | Vivitar (1607225) Telescope
Feb 11, 2009 | Vivitar (1607225) Telescope
Apr 06, 2014 | Nikon Research Microscope Eclipse E800,...
145 people viewed this question
Usually answered in minutes!