Question about Fowler Outside Micrometer Set 0 - 4"
Have you looked at the granite surface plates? I think that it would work really well in this situation. I think that they are sturdy enough that they could get you the stability that you need.
Posted on Jan 18, 2012
My experience in this type of repair spans over 18 years of Calibration experience in working for a Major Automotive Supplier.
So I will attemp to walk you through this. There are several different models of micrometers and each manufacture has a different design. Most of my expertise is with Mitutoyo, Bown&Sharp and Starrett. There could be a couple of other under lying problems due to the "overtightening".
One: could be due to a scratch on the spindle "barrell" from trying to rotate the spindle when it was locked.
Two: the spindle could actually be bent.
Three: If left in the lock position for a period of time, it could have a slight bit of corrosion on the spindle.
If at all possible, remove the lock, loosen the spindle and remove the spindle completely. Place the spindle on a known flat surface. If you have a granite surface plate that would be perfect, if not try a smooth glass top. The idea is to check the spindle to see if it is bent by rolling it and checking for straightness. If it is bent or badly scratched, then it would be cheaper to just purchase a good micrometer from eBay than to try and get parts.
If it is not bent then it could be corrosion so clean with a pure grade alcohol and reassemble.
If you still have the tightness, then it could also be an alignment issue and that is just going to take a lot of patience, and it still may be cheaper in the long run to replace it.
Most precision mics will not tolerate harsh treatment or environments, so be careful..........hope this helps....
Posted on Dec 11, 2008
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 08, 2015 | Fowler Measuring Tools & Sensors
Apr 30, 2011 | Mitutoyo 193 211 Digital Micrometer
Mar 08, 2011 | Fowler Outside Micrometer Set 0 - 4"
Jun 04, 2010 | Mitutoyo #117 107 0 1 Uni Mike Micrometer...
Mar 31, 2010 | Mitutoyo Measuring Tools & Sensors
Mar 28, 2010 | Mitutoyo 0 1".001 Fric/Thimble Mti Outside...
Oct 15, 2009 | Starrett 50443 12 Machinists' Improved...
Mar 07, 2009 | Starrett T216XRL-1 - MICROMETER, DIGITAL
Nov 14, 2009 | Fowler Outside Micrometer Set 0 - 4"
303 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: