NCR 7875 SCANNER/SCALE
Well regarding this error sending you this display of the scanner in most cases is needed because if the scanner calibration, you have to do is:
Remove the dish from scanner (where you put the products to weigh), there you will see that has a plastic part on the right side, and take away that the bottom is a black button color you need to start to put pressure gauge.
Once you press this button, and is ready to start calibrating, then put him back to the plate and the scanner with a voice that sends you will be asking some weight, so this should have on hand your weights to calibrate the scale.
First the scanner asks 0 kg. in this step not going to put anything in weight, ahi press the button Scale Zero appearing in the panel's scanner,
After asks you to 2.5 kg, is there when you need to put the 2.5 kg asks you, when you have them placed on the plate press Scale Zero, the scanner then will ask now that you put 5kg, then you take away those who had 2.5kg and you put the 5kg, and when you have 5kg in the dish press the button Scale Zero, now the scanner will ask that you put 10k, then take away the 5kg and that he had put the 10k that asks you to press the button and Scale Zero, and finally asks you to the scanner in this 0Kg, then you take away the 10k and leave it with nothing, and press Scale Zero
If the calibration did well Scanner you send a voice message that says Calibration Complet or calbración completed, it is very important that when you calibrating the scanner is not moving or having obstacles that may make the calibration is not made well, good clean plate and that this part of the bottom, which revises the 4 Gomito that has to sustain the plate are complete, that are not broken up or are some more than others and that when you put the dish did not move. fijate also not hit the plate or the chassis of the scanner with something else, so there's no obstacles in the calibration,
If you salio calibration is not well in the display will show a 9Error, this means that there was interference during calibration, it is required to check well before the scanner or do a cleanup.
Feb 21, 2008 |