Question about Meade LX200GPS Telescope
After getting a two star alignment (as usual) my scope no longer tracks. I used the scope the night before and had no problem. Please advise, thanks.
Check the time clock battery - it is behind the front panel. If it is dead, then the clock is off, and the alignment will not complete properly. Regular watch lithium coin cell.
Posted on Jul 15, 2008
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Nov 27, 2013 | Meade 10" LX200GPS Schmidt-Cassegrain...
Sep 18, 2011 | Meade 12 Inch LX200R f/10 Advanced...
Jun 02, 2011 | Meade 12 Inch LX200ACF Advanced Coma Free...
Mar 10, 2011 | Tasco Galaxsee 46114375 (375 x 114mm)...
Oct 13, 2010 | Meade LX200GPS Telescope
Jul 02, 2010 | Optics
Jun 04, 2010 | Meade LX200GPS Telescope
Oct 20, 2008 | Meade LX200GPS Telescope
Sep 08, 2008 | Celestron NexStar 8i (600 x 203mm)...
Oct 16, 2012 | Meade LX200GPS Telescope
Sep 09, 2012 | Meade LX200GPS Telescope
171 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: