Question about Kitchen Ranges
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: I have a Avery Berkel
this means the loadcell (weight sensor) is damages and needs replacing,this is best done by an avery dealer as you ned to know how to set up the calibration and trade approval,bob
Posted on Mar 22, 2011
Try these links
www.posguys.com/download/pos6710_l.pdf - United States
Posted on Feb 10, 2012
The zero capture is out of the 2percent tolerance so the cell is either faulty or requires zeroing. Open the unit and press the micros itch on the left of the board. The display flashes s. Press the zero key to change to flashing E. Press store to scroll to zero balance. Press semi auto tare. This captures zero without clearing the span calibration. The display will scroll around for a short period when complete press store.. Release the micro switch and scale will go thru warm up routine then display zero. Hope this helps, regs Darren.
Posted on Apr 15, 2012
Tips for a great answer:
Jul 20, 2017 | Health & Beauty
Jun 02, 2015 | Health & Beauty
Mar 01, 2013 | Health & Beauty
Mar 01, 2013 | Siltec Health & Beauty
Feb 16, 2011 | Avery WeighTronix Berkel WeighTronix 6710 ...
346 people viewed this question
Usually answered in minutes!