Question about Detecto 475 Beam Chair Scale
Make sure the chair is clean and there is nothing between the chair the sensors. Put the setting to zero. if it the arrow goes to center you are good. if not you have to adjust the screw on the other end of the beam. Make a quarter turn at a time. Don't go crazy with it. Try to get the arrow to zero as close as possible. push down on the chair a few times times to make sure it is consistently gets to zero. Get several different weights to check the linearity, 10 should equal 10. 20 should equal 20 and so forth.
Posted on Nov 11, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jun 24, 2015 | Health & Beauty
Oct 30, 2014 | Salter 3007 Aquatronic Digital Scale
Mar 06, 2014 | 760Li Brief instructions f01, f02, f03....
Aug 23, 2011 | Salter 2001 electronic kitchen scale
Jun 15, 2011 | Seca 780 Digital Adult Scale, 440lbs/200kg
May 28, 2011 | Kitchen Appliances - Others
May 19, 2010 | Seca 354 Digital Baby Scale Pediatric...
May 19, 2010 | Robinair Cool Tech 34700Z Recovery,...
Apr 10, 2010 | Health & Beauty
Aug 24, 2009 | Tanita 1479V - Digital Pocket Scale /...
485 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: