Question about Health & Beauty
Marilyn Not having worked on many M6 chairs in a long time I would venture that you have a left motor error indicated by the 3 flashes however since you don't have a indication of full power you might have a battery problem because a low battery can cause the chair to flash also. If you can get someone with a digital voltmeter to measure the voltage at the charger port on the joystick it will let you elimitate the batteries as the problem. Place the voltmeter to 200 volts dc and plug the leads into the two outter holes in the charger port the reading should be 24-27+ volts if the batteries are fully charged. If you are not getting anything or a very low reading then that could be the problem. Replace both batteries with U1 batteries. After yo replace the batteries and charge them before use then turn the chair on and see if you are getting the smae flashes If you are then I suspect it is the left motor and it will need to be replaced,\. Hope this helps you
Posted on Nov 09, 2010
Testimonial: "thank you, I'll try this"
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Mar 24, 2016 | Invacare Pronto M51 Power Wheelchair
Dec 07, 2015 | Invacare Health & Beauty
Nov 24, 2015 | Invacare Pronto M51 Power Wheelchair
Aug 12, 2014 | Invacare Supply Group Pronto M61 Standard...
Jun 22, 2014 | Invacare Supply Group Invacare Pronto M91...
Sep 25, 2012 | Invacare Pronto M51 Power Wheelchair
Sep 16, 2012 | Invacare Supply Group Invacare Pronto M51...
Apr 12, 2011 | Invacare Supply Group Pronto M61 Standard...
Jul 01, 2010 | Invacare Supply Group Invacare Pronto M91...
Aug 27, 2009 | Invacare Supply Group Invacare Pronto M91...
365 people viewed this question
Usually answered in minutes!