Question about Pride Mobility Products Jazzy Select
Yep you have a weak or bad battery. Even tho the chair looks new if they didn't take care of the batteries ie: charge regularly then the batteries went bad just sitting. I would check in your area for a battery warehouse or store that can get you 2 new batteries at a reasonable price. If there is an Interstate battery distributor or store near you the get your batteries there. Your Select uses 2 U1 batteries. Check out my tips on charging your batteries once you have the new ones put in. Hope this helps
Posted on Oct 24, 2010
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Jun 14, 2018 | Select Health & Beauty
Jul 18, 2017 | Select Health & Beauty
Jan 08, 2014 | Pride Jazzy Select 6 Indoor Power...
Oct 22, 2013 | Pride Jazzy 1103 Ultra Standard Power...
Oct 17, 2013 | Pride Jazzy 1113 Power Chair
Feb 19, 2011 | Pride New Jazzy 600 Power Wheelchair by...
Sep 20, 2010 | Pride Mobility Products Jazzy Select 14
Sep 11, 2010 | Pride Jazzy Select 6 Indoor Power...
May 16, 2009 | Pride Mobility Products Jazzy Select
306 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: