Question about Invacare Supply Group Invacare Pronto M91 Electric Wheelchair
With the M91 the click you are hearing is the electric brake solenoid engaging and disengaging the electro mechanical brake. The display blinking probably indicates a brake solenoid fault code and right or left motor. Try working both brake levers back and forth listening for and feeling for smooth action. If the levers are bent and the brakes are not working a motor replacement is necessary. If you can identify the fault flash code you can determine which brake or motor is the culprit and use a DVM to verify the component failure.
Posted on Oct 24, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
May 28, 2014 | Electric Mobility Invacare Pronto M41...
Apr 28, 2013 | Invacare Pronto M91 with SureStep
Mar 16, 2011 | Invacare Supply Group Invacare Pronto M91...
Jan 11, 2011 | Invacare Health & Beauty
Oct 18, 2010 | Invacare Supply Group Invacare Pronto M91...
Jul 30, 2010 | Invacare Supply Group Invacare Pronto M91...
Nov 05, 2009 | Invacare Pronto M51 Power Wheelchair
Jun 19, 2009 | Electric Jazzy 1115 Power Wheelchair
202 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: