Question about ViewSonic Optiquest Q19wb LCD Monitor
Considering the slow discharge of the cap, it is probably good and without an RCL meter with loss capabilities, it would be tough to condemn it.
In most cases I've seen, the larger filter caps are not at fault but rather caps in the range of 4.7-100uFd that are used in the feedback path. The voltage ratings are seldom over 50VDC. These are exposed to high frequencies and if a standard electrolytic is used (they are cheaper) they will eventually fail due to the decay of the chemicals in them.
You might check for a radio amateur club in your area since one of the members might well have an RCL bridge/meter with DF capabilities. If you find one with more than one frequency range, a bad cap will show a major decrease in capacitance and increased losses at the higher frequency.
Posted on Oct 06, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Oct 03, 2010 | Samsung SyncMaster 940N Flat panel Monitor
Mar 28, 2010 | ViewSonic Optiquest Q19wb LCD Monitor
Mar 17, 2010 | Xerox XG-91D 19" LCD Monitor
Apr 03, 2009 | Samsung SyncMaster 931B 19" LCD Monitor
Mar 02, 2009 | ViewSonic Optiquest Q7b Flat panel TFT 17"...
May 19, 2008 | Computers & Internet
May 04, 2008 | Princeton Digital VL2018W Monitor
Sep 17, 2013 | ViewSonic Optiquest Q19wb LCD Monitor
Nov 06, 2011 | ViewSonic Optiquest Q19wb LCD Monitor
106 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: