Question about Premiere Microscopes MF-02 Binocular Microscope
Hi-- I'm having a problem with my MF-02 Binocular microscope stage. It is unstable and will readily "clunk" and move to the left. I've tried tightening various knobs and screws but nothing has helped. Any advice? Thanks!
I think your problem lies in the way the stage is limited upwards. On top of the stage, right behind it where it slides against the arm, there should be a little screw. This screw limits the movement of the stage to a certain point upwards. If the limit exceeds the adjustment gear and its opposing track, then the stage just goes up one notch and then it clunks down. If you screw it down the stage is limited further down, if you screw it up it's limited a bit higher. The point of this screw is to prevent objective and slide damage. Try adjusting this little screw a bit further down and see if it still 'clunks' when you turn it up. After that we may work on properly adjusting it. If not, please post again with whatever detail you can provide to aid you with this problem further.
Posted on Jun 09, 2008
a 6ya Expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to an Expert (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 27, 2011 | Swift M4002D Binocular Microscope
Jan 20, 2011 | Premiere Microscopes MF-02 Binocular...
Jan 16, 2011 | Premiere Microscopes MF-02 Binocular...
Dec 09, 2009 | Motic DMBA 300 Binocular Microscope
Nov 10, 2009 | Advanced Electronic Support Baytronix ...
Nov 25, 2008 | Nikon 7314 Binocular Microscope
Nov 20, 2008 | Premiere Microscopes MF-02 Binocular...
Sep 29, 2017 | Science & Laboratory
180 people viewed this question
Usually answered in minutes!
Level 3 Expert
Level 2 Expert
Level 3 Expert
Step 2: Please assign your manual to a product: