Question about Soehnle Slim F5 Scale 63721
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: scale appears to be inaccurate
this type of less resoluted scale hasn't repeatability. if you wish to repair it, change the loadcell. use 2 x scale capacity of load cell.
Posted on Dec 02, 2008
This is a reply I received from Taylor customer support. I have followed this procedure enough times to wear out the original battery and still get the Err 1 message. You may have better luck.
Thank you for your e-mail. The reboot below may clear the error from
your scale. You may need to do this several times to completely reboot
the system. Please let us know if you have any questions or continue to
display the Error.
1. Remove the battery from the scale
2. Set the scale on the floor without the battery in place.
3. Stand on the scale for 10 seconds (still without the battery)
4. Carefully replace the battery without turning the scale on.
5. Using only one foot, tap the scale or apply just enough pressure to
turn on the display. (It should show dashes or zero's) IMMEDIATELY
REMOVE YOUR FOOT.
6. Let the scale turn off
7. Step on the scale for a weight reading
Taylor Precision Products
& Salter Housewares
2220 Entrada del Sol, Suite A
Las Cruces, NM 88001
Toll free - 800-289-0944
Direct - 575-526-0944 ext 3172
Posted on Jun 24, 2009
I had the same problem with my Salter 987 scale. Pull the battery out for about 10 seconds and put it back in. Then push the scale once so it turns on. After the blinking stops, push it once more. The scale will now zero itself. After the scale has zero'd itself it may turn off after a brief moment. It's now ready to use normally. If you're standing on the scale when it's trying to calibrate to zero, it will assume your weight is zero. Don't stand on the scale after replacing the battery untill you first complete the steps listed above.
Posted on Aug 29, 2010
SOURCE: I recently purchased a Soehnle
Look around and see if it has a reset button ...and reset it to factory settings ..Probably happened with it ..or cause it repeated you are doing something wrong when you set it .....probably you lock it in the last read ..cause it indicates the same measurement ..So ..all devices like that have to have a reset button ...locate it ..(it may be in the menu as a feature..and called different probably ..factory defaults or factory settings ..or reset all ..) and after you engage the reset it will work again ..but ..read the manual carefully and see what are you doing wrong ...
Posted on Mar 21, 2011
Tips for a great answer:
Mar 09, 2015 | Salter 9037 - Digital Bathroom Scales /...
Sep 10, 2011 | Terraillon FBC Alteo Family Body Control...
Jan 05, 2011 | Conair WW81 Weight Watchers Body...
Aug 15, 2010 | Taylor & Digital Scale Lithium Scale
May 14, 2010 | Salter 2001 electronic kitchen scale
Jan 15, 2010 | Health & Beauty
Jan 03, 2010 | Taylor Dgtl Body Fat Scale
Nov 16, 2009 | Conair Weight Watchers WW35 Body Fat Glass...
Jun 13, 2009 | Taylor Body Fat Analyzer & Scale, Model...
Jan 24, 2009 | Health & Beauty
51 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: