Question about Agfa Arcus II Flatbed Scanner
I keep getting a calibration error on my Agfa Arcus II scanner. We have replaced the bulb on the transparency option and it worked a couple of times, but it is giving me the message again. Do you have any solution? The calibration error on the bottom of the warning was 82200000:5
Try disconnecting the transparency module and using the reflective unit only. If that works, the transparency module's connection may be dirty and interfere the calibration process.
If the problem occurred even without the Transparency Module, check those wire connections to the logic board inside the scanner (the ones that can be disconnected without removing the logic board's metal cover). Some of them may become loose. Try this before changing the lamp tube because it's easier, costs nothing and most likely is the culprit.
Posted on Jun 06, 2015
Posted on May 30, 2008
a 6ya Technician can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to a Technician (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
May 01, 2011 | Agfa Arcus 1200 Flatbed Scanner
Jan 19, 2008 | Agfa DuoScan T1200 Flatbed Scanner
Feb 10, 2011 | Agfa Arcus II Flatbed Scanner
Sep 24, 2009 | Sharp AL-1655CS Copier
Apr 24, 2008 | Agfa Arcus II Flatbed Scanner
Nov 16, 2007 | Agfa Arcus II Flatbed Scanner
448 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: