Question about Omron Hj 720ITC Advanced Pocket Pedometer
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
I contacted the manufacturer here is the response with information to contact them yourself if this does not help.
I am sorry to hear that you are having this issue with your readings and would be happy to assist you in determining where the problem lies.
Initially, I would like to provide you with some basic user information on factors that could impact the quality of your readings enough to cause the E error message you have been receiving. I will follow that with some critical steps that you must take in order to obtain a reading on your Omron unit.
User technique is important for reliable measurement results. Listed below are the common causes why you may get an erroneous reading:
Cuff Size - Measure the circumference of your wrist. If the incorrect cuff is used, you may get an inaccurate reading or error messages, if not at the first reading, then after a short period of use because the cuff bladder will be damaged. To determine your wrist size, measure your wrist approximately ½” above your wrist bone using a cloth tape measure. Wrap the tape measure evenly around your wrist. Do not pull the tape tight.
*The cuff that came with your unit is appropriate for wrist circumferences of 5 ¼ to 8 ½ inches. If your wrist circumference is not within that range, your readings will be inaccurate or you may receive error messages.
Cuff Application - Make sure you are wrapping the cuff around your wrist in the correct position. The cuff should be wrapped snugly around your wrist. If your wrist monitor can be used on either the left or right wrist, make sure you have set the monitor to the wrist you are using.
*If the cuff is wrapped too tightly or too loosely, your readings will be inaccurate or you could receive an error message.
Before Taking a Measurement - Avoid eating, drinking alcohol, smoking, exercising, and bathing for 30 minutes and rest for 15 before starting the measurement. Avoid taking a measurement during stressful times. Take the measurement in a quiet place.
Body Position - Sit in a chair with your feet flat on the floor. If you do not have your wrist at heart level, your reading will be affected. Holding your arms or shoulders tensely may affect the reading. Remain quiet while taking the measurement.
The E error message is telling us that the unit is recognizing some type of interference while taking the reading. This could be as simple as speaking or moving, having a computer, cell phone or medic alert device too close to the phone, a loud television/radio in the same room or more complicated, such as a rapid or weak arterial pulse.
Reading Timing - Because Omron units read your blood pressure by 'listening' to the blood as it vibrates through the artery, the artery must be fully opened for an accurate reading. It is critical to wait a full 5-10 minutes between readings to allow the brachial artery to 'bounce back' from the initial constriction.
Refer to the Instruction Manual provided with the unit for the complete operating instructions.
If after reviewing the information above your unit is still not functioning properly, in order to best serve you, please call our Consumer Support team, toll-free at 800-634-4350. Our trained representatives are available Monday through Friday, 830a to 430p CST to assist you over the phone and try to resolve your issue without sending your unit in for repair. If repair is necessary, they can set up your file to start the process or provide valuable options to your particular situation.
Please have the following information available in order to expedite your call:
1. Model number (from the front of the unit)
2. Serial number (from the sticker on the side of the unit)
3. Proof of purchase information - from the receipt or online verification.
4. The wrist circumferences of all users of your unit.
They will be happy to assist you with your options regarding service.
We appreciate your interest in Omron products and want to ensure your satisfaction.
Please let me know if I may be of further assistance and have a wonderful day!
Customer Service Representative
Omron Healthcare, Inc.
Posted on Mar 17, 2009
this website is the Omron site that can let you download the manual to your blood pressure monitor
Posted on Apr 29, 2009
this is the website for Omron Healthcare product manuals where you can find your model number and download or view the manual and even print it out if you want. good luck and God bless!!
Posted on Apr 29, 2009
Tips for a great answer:
Dec 29, 2014 | Omron Health & Beauty
Sep 23, 2014 | Omron HEM 650 Wrist Blood Pressure Monitor
May 02, 2014 | Omron HEM 780 Automatic Blood Pressure...
Oct 14, 2011 | Health & Beauty
Aug 01, 2011 | Omron Automatic Blood Pressure Monitor w/...
Jul 03, 2009 | Omron Automatic Blood Pressure Monitor w/...
Apr 18, 2009 | Omron HEM 780 Automatic Blood Pressure...
Mar 16, 2009 | Omron Automatic Blood Pressure Monitor w/...
118 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: