Question about Omron Model HEM 432C Manual Inflation Blood Pressure Monitor
What is the pattern of the EE? If the two E's are on the same line, then you didn't inflate the cuff to a sufficient pressure. (Wait 2-3 minutes before trying again and then inflate the cuff to 40 mmHg higher than the previous attempt.)
If the Es are arranged vertically, then you overinflated the cuff. Any [ressure above 300 mmHg is beyond the range that this bp monitor can handle. If you reached this point by getting several underinflation alerts, ask your medical professional to test your bp and recommend the starting inflation level that you need.
Make sure that you have the cuff on properly. You may need to consider a different bp pressure cuff size to get an accurate reading.
I hope this.
Posted on Sep 09, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 20, 2014 | Health & Beauty
Jun 28, 2013 | Health & Beauty
Nov 12, 2012 | 1996 Chrysler Town & Country
Aug 11, 2011 | Heating & Cooling
Feb 16, 2011 | Timex Ironman 30 Lap 53952 Wrist Watch
Oct 04, 2010 | Rinnai Toyostove Direct Vent Heater: Laser...
Mar 23, 2010 | Asrock G41M-GS Core 2 Quad/ Intel G41/...
Dec 15, 2009 | Omron Hem 712CLCN2 Cuff Blood Pressure...
697 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: