Question about Ge Medical Systems Ge Marquette Mac 1200 Interpretive Ecg
I would need a little more info to be more exact but there are a few usual causes for this issue. Bad feed roller (you can here it spinning but paper does not move), bad optical sensor (a faint click, very little paper movment), wrong paper (same as bad optical sensor or might just continue to cycle through paper without printing), bad print head (system acts just like it does when it is printing normally but nothing appears on the page) or bad battery (system tries to print but shuts down almost immediatly even if it is plugged in). You can contact me and give me more info and I would be happy to help.
email@example.com or www.overstockme.com or follow us on twitter at overstockmecom
Posted on Sep 04, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
The heat on the print head can fade over time giving you a poor print. also the feed roller can wear and give poor pressure to the print head. thw way to diagnose it is to put a peice of electrical tape on the print roller and print an ekg. if it prints well in the area where the tape was, it is the print roller. if there is no change it is the print head.
firstname.lastname@example.org. www.overstockme.com. Jeff Watson
Posted on Jan 01, 2010
SOURCE: EKG Machine
3 lead machines are designed to monitor the cardiac rhythm cycle, in 3 of the 12 primary leads. An attempt to use this limited information as a true diagnostic tool would be incomplete and perhaps misleading.
A full 12-lead machine is diagnostic, and looks at the elecrical information from all sides ("walls", ie anterior, septal, inferior, lateral, posterior) of the heart and gives detailed information regarding ischemia and or infarction of the cardiac muscle, electrical conduction system function, electrical axis, chamber enlargement, drug effects, and more.
Hope this helps.
Posted on May 08, 2009
You should purchase a printer kit. This is a common problem I have encoutntered on this machine and this machine is identical to the Seca CT6I. I have ordered the printer kit from Seca in the UK, the part number is 9740490024 at a cost of £70. It is a straightforward repair.
Posted on Sep 11, 2009
Tips for a great answer:
Dec 21, 2015 | Medical
Aug 23, 2012 | Schiller Welch Allyn Cardiovit At 2...
Jun 25, 2012 | Schiller Welch Allyn Cardiovit At 2...
Feb 26, 2011 | Schiller Welch Allyn Cardiovit At 2...
Nov 17, 2009 | Ge Medical Systems Ge Marquette Mac 1200...
Sep 07, 2009 | Schiller Welch Allyn Cardiovit At 2...
Jun 10, 2009 | Schiller Welch Allyn Cardiovit At 2...
Apr 07, 2009 | RTO Schiller AT-2i Electrocardiograph -...
Apr 05, 2009 | Schiller Welch Allyn Cardiovit At 2...
Jan 29, 2015 | Ge Medical Systems Ge Marquette Mac 1200...
1,050 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: