Question about Omron Healthcare Health & Beauty
I'm assuming you are using it the way its manual recommends - in the correct position on the arm, the band firmly attached - and that the batteries are new and the contacts clean.
In that case the error appears either because the bladder doesn't inflates or the detector is faulty - in either way it will cost you more to fix it than to buy a new unit. But since this unit is destined to the scrapheap you can try to fix it yourself: open the casing and check that the pump works, that the air pipes leading to the band are in good order (no leaks), that the wires to the sensor on the band are continuous and that the sensor itself is still attached to the wires. If not then try to fix what is broken.
Posted on Aug 31, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 23, 2014 | Omron HEM 650 Wrist Blood Pressure Monitor
Aug 23, 2012 | Omron Healthcare Blood Pressure Monitor...
May 17, 2012 | Health & Beauty
Jan 19, 2010 | Omron Automatic Blood Pressure Monitor...
Dec 15, 2009 | Omron Hem 712CLCN2 Cuff Blood Pressure...
Dec 04, 2009 | Timex Health & Beauty
Apr 18, 2009 | Omron HEM 780 Automatic Blood Pressure...
Nov 22, 2012 | Omron Healthcare Health & Beauty
169 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: