Question about Meade LX200GPS 10 in. Telescope
Hi, I have a serious problem with my telescope Meade LX200 Classic.
I would ask you some opinion about my problem.
Today, when I turn on my LX200 the C1 and maybe C2 capacitors are burning.
I don't really know the reason why.
I hope that the problem is only on C1 and C2 capacitors, but I'm not sure.
Looking at the Power Panel schematic on http://www.ccdguy.com/lx200_images/PowerPanelSch.pdf it seems that the C1 and C2 capacitors are supplied from a +12V that arrives from the Keypad.
It is possible that the real problem is the Keypad?
Sorry for my english.
I hope that yuo can help me.
C1 and C2 are on the input of a voltage regulator. The fact that they burn, could mean that their voltage rating has been exceeded (check the voltage across them), or that age has caused their failure. The classic uses about 18 volts into the power panel - see that it is not higher than that. The handpad would not be a cause for this to happen as it is not the source of the power. Replace the caps with similar ones, but with higher voltage rating - 50v is fine. But check your power supply first!
Posted on May 22, 2008
a 6ya Expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to an Expert (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Feb 26, 2017 | Meade Optics
Feb 15, 2017 | Meade Optics
Jul 14, 2011 | Meade LX200GPS 10 in. Telescope
Mar 17, 2011 | Meade LX200GPS-SMT 8 inch Telescope
May 04, 2008 | Meade LX200GPS Telescope
473 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: