Question about NCR 7875
a 6ya Technician can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to a Technician (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
You would need a PC to communicate it with. Either Via USB or the DataRX port.
Capital Computer Group, LLC
We specialize in the NCR arena, also have many other point of sale needs.
Free bench testing equipment failure on any p.o.s hardware item.
Posted on Apr 22, 2010
SOURCE: NCR 7875 SCANNER/SCALE
First try cleaning under the load plate, and hit the zero scale button. If it doesn't then you need to hit the reset button and recalibrate the scale with weights.
Posted on Mar 13, 2008
Tips for a great answer:
Aug 22, 2016 | NCR 7875-2000 Scanner Scale rs232 I/f
Oct 07, 2011 | NCR 7875
Apr 03, 2010 | NCR 7875
Mar 18, 2010 | NCR 7875
Mar 26, 2009 | NCR 7875
Feb 18, 2009 | NCR 7875
Dec 29, 2008 | NCR 7875
Dec 23, 2008 | NCR 7875
Dec 04, 2008 | NCR 7875
Jun 26, 2014 | NCR 7875
May 28, 2014 | NCR 7875
Jan 30, 2014 | NCR 7875
1,483 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: