Question about Telescopes
The ETX 80 is very difficult to work on. Getting at the flip mirror requires an almost complete teardown of the telescope and its mount. If you are not experienced at repairing the ETX scopes I would not advise trying this youself. I doubt that leaving the scope in the case too long is the cause of the coating damage. It takes years for fungus to really destroy coatings. I have seen newer Meade mirrors losing their coatings prematurely and feel this is a manufacturing defect. If the scope is in warranty it should go back to Meade. If not I'd have my dealer look at it and see if Meae will make an adjustment on it.
Posted on Jun 27, 2008
Tips for a great answer:
Dec 03, 2014 | Meade Telescopes
Sep 12, 2011 | Meade ETX-90EC (325 x 90mm) Telescope
Jul 30, 2011 | Meade RCX400 20 in. Telescope
Feb 02, 2011 | Meade Telescopes
May 16, 2010 | Meade 76AZ-AD (180 x 76mm) Telescope
Mar 05, 2010 | Meade DS-2114 ATS (325 x 114mm) Telescope
Jan 29, 2010 | Meade StarQuest 60AZ-DE Telescope
Dec 17, 2009 | Telescopes
Aug 10, 2009 | Meade DS-2114 ATS (325 x 114mm) Telescope
Jun 12, 2009 | Meade Telescopes
136 people viewed this question
Usually answered in minutes!