Question about Swift M2251C Monocular Microscope
After focusing the microscope the stage starts to slide down slowly. I don't know if this is caused by the adjustment knob or gears in the arm handle. thank you
There is a tension adjustment for this. It is located on the coarse focus knob which is on the same side as the switch. It looks like a chrome ring with about 3 holes in it. A strange looking tool should have come with the scope. It is designed to fit into these holes and allow you to adjust the tension on the coarse focus knob. You need to tighten it just a little bit. Not too much or you will have trouble operating the coarse focus knobs smoothly.
Posted on Dec 25, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
May 16, 2015 | Science & Laboratory
Jan 25, 2015 | Leitz Laborlux K Microscope Wetzlar 513...
Dec 19, 2010 | Tasco 750 Monocular Microscope
Oct 22, 2010 | Swift M2251C Monocular Microscope
Oct 05, 2010 | Premiere Microscopes P-01310 Monocular...
Sep 17, 2010 | Zhumell LOMO P-111 modular brightfield...
Jul 25, 2010 | Premiere Microscopes MRJ-03 Microscope
Mar 05, 2009 | Science & Laboratory
May 28, 2008 | Premiere Microscopes MF-02 Binocular...
335 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: