Question about Walgreens Blood Pressure And Pulse Monitor
Hi.The reason for the EE code is generally caused when the cuff has been wrapped too loose around the arm. Essentially,your cuff was under-inflated during the measurement procedure. This error code will also display if the cuff is worn and needs to be replaced. The EE symbol with a down arrow indicates the cuff is over-inflated. In most cases the cuff has been wrapped too tight or you are using the incorrect cuff size. Try Turning the monitor off and pulling the batteries(if equipped); for five to ten minutes then reinsert the batteries and reset. Read the instructions to correctly apply the cuff. If the unit continues to show the EE code, you will need to purchase another cuff.
Posted on Jul 06, 2010
a 6ya Expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to an Expert (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Oct 06, 2017 | The Health & Beauty
May 08, 2016 | Omron hem-432c manual arm digital blood...
Nov 06, 2017 | Walgreens Blood Pressure And Pulse Monitor
Aug 05, 2011 | Walgreens Automatic Wrist Blood Pressure...
May 16, 2011 | Health & Beauty
Sep 21, 2010 | Walgreens Blood Pressure And Pulse Monitor
Sep 09, 2010 | Omron Model HEM 432C Manual Inflation...
Aug 19, 2010 | Omron Model HEM 432C Manual Inflation...
Dec 15, 2009 | Omron Hem 712CLCN2 Cuff Blood Pressure...
Feb 21, 2009 | Omron Automatic Blood Pressure Monitor...
1,375 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: