Question about Invacare Supply Group Invacare Scooter Lynx L-3 L-4 Transaxle Differential
This could be caused by a couple of different issues. If your batteries fall to below 17 volts the unit will shut down. After it sets for a few minutes it may run again for a short time. If this is the case, then the batteries are not being charged properly or they are not holding charge.
Another cause could be the break switch circuit is opening. If that is the case, replace the brake/motor harness
Posted on Jun 06, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jun 16, 2016 | Electric Strider Maxi 4 Wheel Mobility...
Mar 26, 2015 | Health & Beauty
Apr 21, 2017 | Health & Beauty
Jun 20, 2014 | Pride Celebrity X 4-Wheel Electric Scooter...
Dec 31, 2013 | Lynx L-4 Motorized Scooter
Dec 29, 2013 | Motorcycles
May 02, 2011 | Health & Beauty
Aug 28, 2010 | Health & Beauty
Oct 31, 2009 | Electric Strider Maxi 4 Wheel Mobility...
Aug 10, 2014 | Invacare Supply Group Invacare Scooter...
Mar 11, 2017 | Invacare Supply Group Invacare Scooter...
274 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: