Question about Mitutoyo #117 107 0 1 Uni Mike Micrometer .0001
This is based on an old Brown & Sharp guide book. 1. Carefully clean the measuring surface by pulling a piece of soft paper between the surfaces while the anvil and spindle are in light contact with it. 2. with the anvil and spindle apart, unlock cap with spanner wrench (supplied with unit), then tighten cap lightly with fingers to bring slight tension between thimble and spindle. 3. Bring anvil and spindle together by turning SPINDLE and set zero line on thimble to coincide with line on barrel. 4. Move spindle away from anvil by turning SPINDLE, NOT by turning thimble. 5. Holding thimble only, tighten cap with fingers. Do Not Touch Frame. Lock cap with wrench, still holding thimble only, and adjustment is complete. NOTE: Check zero setting after adjustment is made. Hope this helps
Posted on Jun 08, 2010
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 08, 2015 | Fowler Measuring Tools & Sensors
Apr 03, 2014 | 2004 Yamaha V Star 1100 Silverado
Feb 21, 2013 | 2009 Yamaha XCity 250
Nov 20, 2012 | Mitutoyo 317-351 Digital...
Jun 29, 2011 | Mitutoyo Depth Micrometer 0-6"
May 02, 2010 | 2006 Yamaha Stratoliner
Aug 15, 2009 | Mitutoyo 293 831 Electronic Digital...
Jul 28, 2009 | Mitutoyo Measuring Tools & Sensors
Mar 07, 2009 | Starrett T216XRL-1 - MICROMETER, DIGITAL
Feb 03, 2009 | Health & Beauty
349 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: