Question about Welding Tools
Sartorius PMA 7501 electronic paint mising scales showing error code no. ER -291 on display .It is now not working and or not happning on off
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: NCR 7875 SCANNER/SCALE
Well regarding this error sending you this display of the scanner in most cases is needed because if the scanner calibration, you have to do is:
Remove the dish from scanner (where you put the products to weigh), there you will see that has a plastic part on the right side, and take away that the bottom is a black button color you need to start to put pressure gauge.
Once you press this button, and is ready to start calibrating, then put him back to the plate and the scanner with a voice that sends you will be asking some weight, so this should have on hand your weights to calibrate the scale.
First the scanner asks 0 kg. in this step not going to put anything in weight, ahi press the button Scale Zero appearing in the panel's scanner,
After asks you to 2.5 kg, is there when you need to put the 2.5 kg asks you, when you have them placed on the plate press Scale Zero, the scanner then will ask now that you put 5kg, then you take away those who had 2.5kg and you put the 5kg, and when you have 5kg in the dish press the button Scale Zero, now the scanner will ask that you put 10k, then take away the 5kg and that he had put the 10k that asks you to press the button and Scale Zero, and finally asks you to the scanner in this 0Kg, then you take away the 10k and leave it with nothing, and press Scale Zero
If the calibration did well Scanner you send a voice message that says Calibration Complet or calbración completed, it is very important that when you calibrating the scanner is not moving or having obstacles that may make the calibration is not made well, good clean plate and that this part of the bottom, which revises the 4 Gomito that has to sustain the plate are complete, that are not broken up or are some more than others and that when you put the dish did not move. fijate also not hit the plate or the chassis of the scanner with something else, so there's no obstacles in the calibration,
If you salio calibration is not well in the display will show a 9Error, this means that there was interference during calibration, it is required to check well before the scanner or do a cleanup.
Posted on Jun 30, 2008
Put scale in round file can. Buy another cheap scale.
Posted on Oct 05, 2009
Testimonial: "The new scale works great and was on sale at Wal-mart for only $29."
It's not the scale board. Its an undocumented program setting. Change as follows (you will need programming scan codes from NCR document downloads)
Hex 2 (or Hex 1 for Kg)
Then it will request a calibration procedure (Done)
Posted on Aug 04, 2012
SOURCE: 'out 2' error message shows
Below is a solution for the "Out2" error on the Salter scales.
I have a Salter Model 1400. It was dropped and started to display the "out2" error on the LCD display. I looked for solutions on-line but could not find anything which showed a solution. With no solution in hand I figured I would at least try to find out if my scale was salvageable. I removed the screws, there are two below the 9V battery, two under the footpads, two under the "model sticker", and two larger screws holding the scale plate to the base. After all the screws are successsfull removed, the case of the scale comes apart easily. I started looking at the small circuit board and looked for damage, and also tried several things, turned on and off the unit while it was apart but still saw the "out2" error. I then removed the circuit board, there are many screws holding it down. Looking at the LCD display, I noticed that there must be proper contact between the LCD display and the copper contacts on the circuit board, or nothing is displayed. Perhaps, "out2" is an indication that it can not display "out" anything because after it was dropped the LCD display moved enough that it no longer aligned with the contacts on the circuit board. I noticed that there is quite a bit of "play" or area of movement around the LCD display and the case. There were no cracks or damage so I assume that the loose LCD display is common even with a scale that was not dropped. I was able to align the LCD display, by moving it several times and realized that after pressing the ON button, I no longer saw the out2 error. So I decided to put something in between the LCD display and the case so that the LCD display would not move and therefore would align. It took some playing around, but I eventually took a very small piece of papertowl, about the size of my thumbnail and rolled it up and placed it between the LCD display so it made the LCD display snug and would no longer move around. I screwed on the circuit board, and voila, no more out2 error! It started working perfectly.
Posted on Sep 21, 2012
Tips for a great answer:
Sep 12, 2014 | Salter 2001 electronic kitchen scale
Mar 03, 2011 | Seca 780 Robust Digital Column Scale /...
Feb 14, 2010 | Salter 2001 electronic kitchen scale
Oct 03, 2009 | Dmm Elscale Healthsmart Glass Electronic...
Jul 02, 2009 | Taylor 5757 Digital Glass Body Fat Scale
Apr 22, 2009 | Taylor 5757 Digital Glass Body Fat Scale
405 people viewed this question
Usually answered in minutes!