- If you need clarification, ask it in the comment box above.
- Better answers use proper spelling and grammar.
- Provide details, support with references or personal experience.
Tell us some more! Your answer needs to include more details to help people.You can't post answers that contain an email address.Please enter a valid email address.The email address entered is already associated to an account.Login to postPlease use English characters only.
Tip: The max point reward for answering a question is 15.
Want to calibrate the MSA Altair 4 Gas Detector or MSA Altair 4X? Here's how:
1. Turn the MSA Altair 4 and MSA Altair 4X on in normal reading mode. Do this by pressing and holding the center button.
2. Once the MSA Altair 4 and MSA Altair 4X is in normal reading mode, you need to send it into calibration mode. Do this by pressing and holding right beeps at you and the screen asks you to zero the monitor (sensor refresh on the MSA Altair 4X).
3. Now that the screen is asking for confirmation, press the center button once to confirm. This will send the monitor into zeroing mode. Now is a good time to set the monitor down, and grab your calibration gas, regulator, and calibration adapter.
4. Open the valve on your regulator. This is a very important step so you do not ruin your calibration gas cylinder. Now screw it in to your calibration gas cylinder. Close the regulator once you hear gas spewing out of it. This is the signal to know your regulator is good to close. Continue screwing in the regulator until it stops. Do not overturn it.
5. Attach the tubing to the hose barb on your regulator. Make sure the tubing is also connected to your calibration adapter.
6. Wait for the screen to come up asking for you to apply calibration gas. Once this screen comes up, press the center button again to confirm. Screw the calibration adapter in to your gas detector. Once it is screwed on, turn on your gas.
7. Make sure your calibration gas values on the screen match what is on your calibration gas cylinder!
8. Once the calibration procedure is completed, turn your gas off and remove the calibration adapter from the MSA Altair 4 or MSA Altair 4X. The screen will go back to normal reading mode within a few seconds.
9. Congratulations! You have officially calibrated your MSA Altair 4 or MSA Altair 4X gas monitor. Make sure to record these readings in your calibration log!
If you come up with an error, the most likely bet is a bad oxygen sensor, but it could be a number of other issues. Feel free to email me.
Could you please explain more about your problem. Are you able to get your monitor working or is it that it keeps on blinking without any display on screen.
If your CPU is ON and then your facing no display on the monitor then the problem could not be with the monitor instead it could be either a loose RAM. Also check that the cables attached to the CPU is not loose.