Question about Tasco 750 Monocular Microscope
We have 20 monocular Leica BME scopes. Several of them have had the tube that holds the optic lens separate from the rotatable head of the microscope. What is left is a black fitting with a square glass peice in the bottom. Is this a friction fit or a glued joint? They are very loose and no longer snug up. Would you suggest a glue that can be used or is there a better fix to this problem?
I understand your problem and I would
Recommend stud-fix by Loctite, making as sure as possible that the tube occupies the exact same place as it did before removal (at right angles to the optical axis)
If I can be of further assistance, please do not hesitate to contact me.
Posted on Mar 10, 2008
Sir,your microscope tube with the eye piece is screw joint with microscope body. Rather part is friction fitted as well as there is a proper place to mount it.you can do the gluing but first of all be clear that part should be not a movebale part other wise you will have dificulty for fine focusing.
for the further quotation dont be hazitate,
Posted on Feb 19, 2008
The tube joint was a friction fit that had become loose. Most of the time it is a screw joint which is why I was confused. We put some electrical tape around the inside portion of the fitting and the tube fit snugly down over it. We'll have to wait and see how long it lasts.
Posted on Feb 14, 2008
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Mar 02, 2015 | Swift M3500 Monocular Student Microscope
Dec 11, 2011 | Intel Play QX3 Monocular Microscope
Sep 21, 2011 | Intel Play QX3 Monocular Microscope
Oct 13, 2010 | National Optical 138-MS Monocular...
Jul 28, 2010 | Leica DM E Trinocular Microscope
Feb 28, 2010 | Zeiss Microscope
Oct 06, 2009 | Swift M2251C Monocular Microscope
Jan 23, 2009 | C A Scientific MS01 Monocular Microscope
Sep 11, 2007 | Leica ZOOM2000 Binocular Microscope
Mar 03, 2007 | Meade 9260 Monocular Microscope
119 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: