Question about Meade ETX-90 (48x90 mm)
My telescope has started sticking when slewing left,it stops and makes a clicking noise,can you help
Some (not all) versions have an end stop - it it just reaching that? The scope will rotate only aboit 3 complete turns between the 2 ends.
The 'correct' starting position is as follows: Place scope on tripod or table with control panel at the West. Unlock the declination lock (horizontal slew). Rotate scope anticlockwise until it hits the end stop, Then roate it back clockwise until scope is pointing North (should be about 3/8 of a turn, as the end stop is at South West). Then lock the declination, Set the tube horizontal and continue with the alignment process.
Autostar always thinks you have set the position like that. If you haven't, it will work OK until it tries to rotate past the point it won't go, then chaos results!
Posted on Aug 15, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jun 16, 2013 | GSI Super Quality 20x60-80 Porro Prism...
Feb 23, 2008 | Fujinon Super 80 (25x80 mm)
Feb 01, 2011 | Tasco World Class 20-60x60 (20-60x60 mm)
Mar 25, 2010 | Meade ETX-90 (48x90 mm)
Apr 11, 2009 | Meade ETX-90 (48x90 mm)
Apr 30, 2008 | Meade ETX-90 (48x90 mm)
Apr 24, 2008 | Meade ETX-90 (48x90 mm)
Apr 18, 2008 | Meade ETX-90 (48x90 mm)
Nov 12, 2007 | Meade ETX-90 (48x90 mm)
206 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: