Question about Meade LX200GPS Telescope
I have been using my LX200 10" for several years in Alt/Az position and have had many pleasurable viewing secessions and mediocre imaging secessions. I decided to build a concrete pier mount and Milburn wedge setup to advance my photo experience. I mounted my scope on the wedge and began my alignment process. As soon as the scope started slewing (btw the ra and dec directions are reversed so I had to stop the slewing) and then started slipping. The worm gear must have partially disengaged under the weight of the scope. the only accessories mounted to the scope were my 10x50 viewfinder and a newly added 80mm apo for auto-guiding. No cameras or counterweights were even mounted. I took off the base cover to inspect the gears for any damage do to this slipping. The gears appear to be in decent shape however I did not disassemble and clean the grease off to thoroughly inspect. I do not see any shavings or chips etc. so I believe there is little to no damage. I am suspecting that the spring tension on the worm gear/ drive motor is the culprit. Is there a Meade (or alt source) upgrade for this condition.
Those who always look up, find the wisdom to look beyond...
The problem is the addition of the 80mm telescope to the assembly. The telescope is now out of balance, and places extra weight off-center as the position chages. Too great an off-balance will cause the main gears to 'jump' the worm teeth. The solution is to restore balance by adding counter weights as needed. Loosten the axis locks (carefully) and see which direction it wants to move. To balance my LX, I used a small bag with lead buckshot to determine how much weight was needed and where to achieve balance. Then I added fixed weights to the mount. Once this is done, the scope should track correctly.scopestuff.com has balance kits at reasonable prices.
Posted on May 22, 2008
a 6ya Expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to an Expert (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 18, 2011 | Meade 12 Inch LX200R f/10 Advanced...
Oct 08, 2010 | Meade ETX-125EC (500 x 127mm) Telescope
Jul 18, 2010 | Meade LX200GPS Telescope
Jul 02, 2009 | Meade Infinity 114EQ-DH Telescope
May 16, 2009 | Orion SkyQuest XT8 Dobsonian Telescope
Apr 16, 2009 | Meade LX200GPS Telescope
Nov 17, 2008 | Meade ETX Premier Edition: ETX105 PE...
Oct 17, 2008 | Meade Optics
Sep 14, 2017 | Meade LX200GPS Telescope
Oct 16, 2012 | Meade LX200GPS Telescope
909 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: