Question about Agfa DuoScan T1200 Flatbed Scanner
I just bought a "new" T1200.
I connect it to my PC over a 2906 SCSI-card (I use this for my T2500 as well).
When I tried to do the first "preview" of a negative, I got following error-note:
(translated from Dutch)
There has occured a calibrating-error.
Make sure that the full look-through-surface of the calibrating-opening is free or that the negatif-carrier-frame is properly in place inside the scanner...
What can be the problem...?
Kind regards, Herman.
Hi i have found this link for you see if its any help to you if not let me know and ill have another look ok
Here is the link.
www.bioresearchonline.com/.../news/article.asp?Bucket=Article&DocID=6c5bfe81-1c6c-11d4-8c38-009027de0829 - 28k -
Posted on Feb 12, 2008
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
May 15, 2008 | Agfa DuoScan T1200 Flatbed Scanner
Feb 06, 2013 | Agfa DuoScan T1200 Flatbed Scanner
Mar 08, 2011 | Agfa DuoScan T2500 Flatbed Scanner
Mar 27, 2009 | Agfa DuoScan T1200 Flatbed Scanner
Dec 09, 2008 | Agfa DuoScan T1200 Flatbed Scanner
Feb 08, 2008 | Agfa DuoScan Flatbed Scanner
Jun 06, 2007 | Agfa DuoScan Flatbed Scanner
437 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: