Question about Agfa DuoScan T1200 Flatbed Scanner
I just bought a "new" T1200.
I connect it to my PC over a 2906 SCSI-card (I use this for my T2500 as well).
When I tried to do the first "preview" of a negative, I got following error-note:
(translated from Dutch)
There has occured a calibrating-error.
Make sure that the full look-through-surface of the calibrating-opening is free or that the negatif-carrier-frame is properly in place inside the scanner...
What can be the problem...?
Kind regards, Herman.
Hi i have found this link for you see if its any help to you if not let me know and ill have another look ok
Here is the link.
www.bioresearchonline.com/.../news/article.asp?Bucket=Article&DocID=6c5bfe81-1c6c-11d4-8c38-009027de0829 - 28k -
Posted on Feb 12, 2008
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
May 15, 2008 | Agfa DuoScan T1200 Flatbed Scanner
Feb 06, 2013 | Agfa DuoScan T1200 Flatbed Scanner
Mar 08, 2011 | Agfa DuoScan T2500 Flatbed Scanner
Mar 27, 2009 | Agfa DuoScan T1200 Flatbed Scanner
Dec 09, 2008 | Agfa DuoScan T1200 Flatbed Scanner
Feb 08, 2008 | Agfa DuoScan Flatbed Scanner
Jun 06, 2007 | Agfa DuoScan Flatbed Scanner
430 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: