Question about Boreal WW3111900 Microscope
The problem is that the stage is heavy and the adjustment screws are loose. You need to tighten the coarse adjustment knob screws in order to create more friction so that the stage stays put. That, or don't adjust anything and keep holding on to the coarse adjustment while you're using it.
Posted on Nov 23, 2009
You can remove pieces from the stage like the condenser lense if you don't have the right tools for adjusting the screws
Posted on May 22, 2014
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 30, 2014 | Nikon Eclipse E600-fn Microscope Polarizer...
Jul 27, 2012 | Science & Laboratory
Dec 01, 2011 | Celestron 44345 Microscope
Dec 01, 2010 | National Optical 131 Monocular Microscope
Oct 05, 2010 | Premiere Microscopes P-01310 Monocular...
Jul 25, 2010 | Premiere Microscopes MRJ-03 Microscope
Jul 08, 2010 | Swift M2251C Monocular Microscope
Jan 04, 2010 | Science & Laboratory
269 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: