Question about Omron HEM-711 IntelliSense Automatic Blood Pressure Monitor
Same problem on my Omron HEM-711. Manufactured in 2003, past the 5-yr warranty. Cust Suppt told me to replace unit. I opened the unit up and looked inside. Two very tight phillips screws, one on back panel, other in battery compartment. Case pries apart. LCD display looked intact, ribbon was undamaged. Unit appeared to work properly, just did not display correct reading, some of the liquid crystals in a few of the digits were not activating. I installed batteries with case still open, turned power on while pressing ribbon along edge of circuit board. Ribbon is manufactured with glue to circuit board, some of the leads come loose after several years due to manufacturing defect. In my case, pressing towards the bottom side corrected problem. You can see when it's working properly because all the display crystals light up simultaneously when unit is first powered up. I clamped down ribbon with two small paper clips insulated with a few bits of paper, works fine now, saved eighty bucks.
Posted on Sep 27, 2012
Machine was out of warranty as is mine, which was purchased in 2007 (5 yr). I've found 2 screws underneath mine in the front but none in the battery compartment. Top / bottom will not separate and feels like there must be 2 more screws but don't see them. Anyone know how to get apart without damaging the unit? I have the same issue as above with the led crystal display disappearing. Otherwise, it appears to be functioning correctly.
Posted on Jan 06, 2016
Omron must replace it for you free of charge.
Posted on Sep 06, 2009
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Oct 06, 2016 | Omron HEM-711 IntelliSense Automatic Blood...
Feb 19, 2015 | Omron Health & Beauty
Nov 01, 2014 | Omron Automatic Blood Pressure Monitor w/...
Oct 02, 2014 | Omron HEM-711 IntelliSense Automatic Blood...
Sep 25, 2011 | Omron HEM 711AC Digital Blood Pressure...
Nov 30, 2010 | Omron HEM-711 IntelliSense Automatic Blood...
Oct 11, 2010 | Omron Automatic Blood Pressure Monitor...
Jul 30, 2009 | Omron Healthcare 2007 Hem 711DLX Automatic...
1,460 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: