Question about Schiller Medical
If this is the model i'm used to: remove power then: remove all leads then repower go into menu and, reset then: using one found good lead plug it into the machine jack assuring that each jack recognizes the good lead
then now knowing your lead jacks are recognized check and, verify each additional lead. if any of the reset procedures fail have the machine serviced
Posted on Sep 05, 2009
Internal circuit of the unit needs to be checked if there is a short inside the machine will not detect the leads.also check out the leads in another machine to confirm the faulty part. thanks.please keep updated.please do rate the solution if helped.thank you for using fixya.
Posted on Sep 04, 2009
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Does it work in auto mode ? are you putting paper correctly means , there is a black mark on paper which is sensed by paper sensor .If you put paper up side down then paper sensor won't be able to sense any paper & gives you error for rifilling paper .If this doesn't help you then You need to replace battrey . Thanks
Posted on Jul 22, 2009
Tips for a great answer:
Jul 07, 2017 | Schiller Medical
Sep 28, 2013 | Schiller Cardiovit At 1 Ecg Machine
Sep 12, 2011 | Schiller Welch Allyn Cardiovit At 2...
Jun 23, 2011 | Schiller Welch Allyn Cardiovit At 2...
Jun 14, 2011 | Schiller Welch Allyn Cardiovit At 2...
Mar 31, 2011 | Schiller Welch Allyn Cardiovit At 2...
May 09, 2010 | Schiller Welch Allyn Cardiovit At 2...
Mar 17, 2010 | Schiller Welch Allyn Cardiovit At 2...
Jul 15, 2009 | Schiller Welch Allyn Cardiovit At 2...
Oct 24, 2008 | Schiller Cardiovit At 2 Light Ecg Machine
60 people viewed this question
Usually answered in minutes!